The Rapid Detect 12 Panel CLIA Waived Drug Test Cup is a one-step immunoassay device that uses the principle of competitive binding. It simplifies urine collection and maintains fast and reliable test results. This drug test cup offers a clean and affordable approach to onsite urine drug testing and drugs of abuse detection. The device is equally comfortable to obtain a urine specimen for both males and females due to the round opening at the top of the cup. These cups test for the lowest SAMSHA cut-off levels for greater sensitivity.
Drug Configuration with Cut-offs
Marijuana 50, Cocaine 150, Morphine 300, Methamphetamine 500, Amphetamine 500, Phencyclidine 25, Benzodiazepines 300, Ecstasy 500, Oxycodone 100, Barbiturates 300, Buprenorphine10, Methadone300
- CLIA Waived
- SAMHSA low cut-off levels (AMP 500ng/mL, COC 150 ng/mL, mAMP 500 ng/mL)
- Negative results as soon as 30 seconds
- Strong negative line intensity
- Clearly differentiated results at +/- 50% cutoff
- Up to 99% Accurate
- FDA 510(k) Cleared
- Health Canada Approved
- Compare to iCup
The Rapid Detect cup boasts an accuracy of up to 99%. Drug cut-off levels are set according to SAMHSA standards. Rapid test results for a wide range of available illicit drugs can be selected. This 12 panel CLIA Waived drug test cup displays strong negative line intensity and results are easy to distinguish, negative or positive. Test operation is simple to perform being guided by easy-to-read instructions as well as a diagram provided within the package insert.
A product called
UrinAssist can also be purchased for carefree transport of urine to the cup eliminating spillage & contamination of hands, clothing, container & surroundings.
How to Use the Control Region (C)
A procedural control is included in the test. A colored line appearing in the control line region (C) is considered an internal procedural control. It confirms sufficient specimen volume, adequate membrane wicking, and correct procedural technique. The control will communicate to you whether the device is working properly or not. Once the urine specimen is added to the cup you will want to verify if there is a line in the control region (C).
How to Store Test Cups?
Store the sealed pouch either at room temperature or refrigerated (2-30C). The device is stable through the expiration date printed on the sealed pouch. The test device must remain in the sealed pouch until use. Keep away from direct sunlight, moisture and heat. Do not freeze. Do not use the test devices beyond the expiration date.
When Should I Collect Urine for the Test?
The minimum time is 2-7 hours, so you may collect urine samples 2-7 hours after suspected drug use.
What Materials are Provided with Your Purchase?
- 25 Test Cups sealed in foil pouches.
- Package Insert
- Test Procedure Card
- Disposable gloves
What Materials are Required But Not Provided?
- A timer set to 5 minutes before reading results
- Test Strips for Adulterants